Nobska MAVS-3 3 axis acoustic current meter instrument
Type: Current meter
|METADATA||70||EMSO > Western Ionian Sea||SN1 TEST||1 sample / 10 min||May 16, 2018 - December 31, 2019|
|METADATA||38||EMSO > Western Ionian Sea||SMO 1||2 Hz||June 10, 2012 - June 12, 2013|
|METADATA||3||EMSO > Iberian Margin||NEAREST 1||5 Hz||August 26, 2007 - July 16, 2008|
|storage type||compact flash card|
|memory capacity||128 Mbyte|
|2012-10-23||The current meter has been manually reconfigured to aquire at 2 Hz for the first 20 minutes of every hour, remaining "silent" for the remaining 40 minutes.
It was also setted the correct UTC time.
The configuration is currently under testing and will must be reconfigured in case of poweroff.
The following configuration parameters were selected for the current meter:
Measurement Frequency: 2.0 [Hz]
Measurements/Sample: 1 [M/S]
Sample Period: 0.50 [sec]
Samples/Burst: 2400 [S/B]
Burst Interval: 0 00:00:00 [DDD HH:MM:SS]
At 13.30 UTC The Acquisition Software was restarted.
|2012-10-11||Between 8.00 and 9.45 UTC, current meter tests and acquisition software restarted|
|2012-10-10||Between about 13:55 and 15:05 UTC, tests were executed on changing of sampling rate of the Nobska current meter. At the end, the acquisition software was restarted|
|2012-09-10||At about 09.20 UTC started testing current meter configuration. At 10.55 UTC test completed and restarted acquisition software.|